The span gas portion of the custom calibration may be inaccurate. Set back to factory calibration and see if you get a similar response. If yes, then ensure there are not some high levels of VOCs present. Remove lamp and clean, blow out cell and re-calibrate. Contact your local Authorised Service Centre if this fails.